This item is produced according to ISO/CD8782-I Standard, test grounding resistance of anti-static shoe/heel grounders and wrist straps &with LED display and alarm, it is a measuring instrument that cannot be short of during static control. It adopts 9V battery for power supply, with low voltage but safe. It has alarming function when voltage shortage occurs,height can be adjusted freely.
Test if shoes are conductive,anti-static or insulative,judge grounding resistance of shoes,left LED indicate left shoes,right LED indicate right shoes:
|102-103||FAIL LO||Without alarm, yellow light on|
|103-104||FAIL LO||Without alarm, yellow light on|
|104-105||PASS||Green light on|
|105-106||PASS||Green light on|
|106-107||PASS||Green light on|
|107-108||PASS||Green light on|
|108-109||FAIL HI||With alarm, red light on|
Push the option switch to “WRIST STRAP” on the right, put on wrist straps and then insert the plug or alligator clips into the socket (alligator clips pole) and press “PRESS TO TEST” key. When the RIGHT indicator light is on, it means the impedance of the wrist strap.
Test Wrist Strap:
|R〈750K||FLIL LO||With alarm, yellow light on|
|750K〈R〈10M||PASS||Green light on|
|R 〉10M||FAIL HI||With alarm, red light on|
|Note: Testing precision within +15%, calibrated by decade resistance box.|
How to Fix:
1.Loose sleeve,insert bracket through testing board and screw sleeve,as chart A
2.Fix up connecting wires on the two sides of testing board separately and fasten the bracket on the board.(please well note:2 connecting wires correspond two testing loops,blue wires is for left footwear and black for right )
3.Fasten tester batholith on the bracket and fetch connecting wires out.as chart B.
4.Insert connecting wires into tester jack,as chart C.The blue wire is on the left and black is on the right,as chart D.
How to Adjust Height:
Contrarotate sleeve for several circles and adjust bracket to a good height,then screw the sleeve.
In order to save time,we can use hanging installation.Take tester out and use another set of connecting wires.Install connecting wires according to mark,another ender of wire link to the soleplate.There are two nails with 86mm of distance on the wall.The user can hang the tester on the nails.