This instrument is produced according to ISO/CD8782-1 standard , test ground resistance of hand & feet, there are LED display & alarm. It is an absolutely necessary tester in static control industry. ESDMAN-6506 uses 9V battery , low voltage and safety. User can easily adjust the height of tester.
Remove tester unit from the carton and inspect for shipping damage. Each unit should include the following:
1.-Tester unit. Item # ESDMAN 001-6506
When press the test button ,test results within 5 seconds will be shown on the LCD screen ,if the press time is larger than 5 seconds,testing will be stopped automatically.Do not press the test button heavily.The test procedures are very simple and easy: Operator stand on the footplate with heel grounder or anti-static shoes,connect wrist strap and plug socket or alligator clip into the wrist strap jack,then press”PRESS TO TEST”button,LCD screen will tell you the test results.
Testing OK comes with beep sound and opening gate signal,operator need to push gate to enter.If testing NG need to check footwear or repeat the testing procedure.
A.Slide switch testing range adjust：
B.Testing function adjust
|18.104.22.168||Down||Down||Down||Down||Wrist Strap + Left foot + Right foot|
|22.214.171.124||Down||Down||Up||Down||Left foot / Right foot (one of them)|
|126.96.36.199||Down||Down||Down||Up||Wrist Strap Only|
|188.8.131.52||Up||Up||Up||Down||Left foot + Right foot (Both)|
Note：Accuracy+10%,calibrated by resistance decade box
Left foot / Right foot (one of them):Any of the foot strap is tested pass can give signal opening the gate
Left foot + Right foot (Both):Need both foot straps tested pass to give signal opening the gate.