grounding items. Its unique design embeds an ESD tester, computer clock, touch screen display, IC / ID card reader or
barcode scanner, Ethernet module etc. into a compact stainless steel enclosure.
This system can be worked with tripod turnstile, swing gate, flap gate, electronic lock & air shower room. In addition, it could
provide basic ESD testing management, stop operators entering EPA without ESD testing.
has the function of distinguishing and setting up different attendance statement. It will automatically store 5,000 testing
results while software is closed and PC is off-line.
1. Communication: TCP/IP
2. Touch press button design
3. LCD display screen shows: Time, Date, Position, Department, Employee ID, Name, Group, Tested Item and detailed
resistance of wrist strap and footwear
4. Basic Permission Set Up:
Card / Barcode + Wrist Strap
Card / Barcode + Footwear
Card / Barcode + Wrist Strap + Footwear
Card / Barcode + Wrist Strap or Footwear
5. After swiping the card (or barcode), system will be automatically reset if not going through in limited pass time.
6. ESD Testing Steps:
one of the ways to give the legal permission)
test results are displayed on the touchscreen.
DO NOT touch any other metal while performing your test as this will affect your results.
Notes: When performing a footwear test, be sure to place both feet on the dual foot plate (one foot per plate). Keep the
foot-plate clean with 99% isopropyl alcohol when using the 1 Gigohm high test limit. A dirty foot plate could yield a false
pass. When performing a wrist strap test, be sure to completely plug in the wrist cord into the tester’s jack.
7. IP can automatically reset, default IP is 192.168.1.156, IP address can be changed upon request.
8. 12V Wiegand Interface to external connect card reader.
9. 5V S232 Serial Port to external connect LCD display screen.
Simultaneous testing of the worn wrist strap and foot grounders in less than 2 seconds
Records results for each foot grounder and each conductor in single-wire and dual wire wrist straps
Digitally programmable test limits(online version)
Dual foot plate design independently tests each foot grounder
Capable of automatic reset. When the card is swiped, but no one passes within the access time limit, the system will automatically reset.
The access time limit can be set based on the actual needs as requested.
The alarm will be triggered in case the employees press the button for ESD test before swiping cards and registering identity, which can
remind employees to develop desirable habits and prevent time-consuming congestion.
5-inch Color Touch Screen Display
Single-Wire and Dual-Wire Wrist Strap Testing
Heel Strap and ESD Shoes Testing
IC (ID) Card Reader (or Embedded Barcode Scanner)
Ethernet Module for Network Communication
Dual Footplate for Independent Footwear Testing
Paperless Data Logging VS Traditional Paper Logs
identifying problems while reducing logging and auditing costs.
inaccurate and ignored over time.
Tester Software Features
Records Temperature and Relative Humidity
Records Resistance Measurements
Displays Operators First and Last Name
Displays Operators Test Requirement
Color-Coded Pass / Fail Results
Access Control Capabilities
Records In and Out Transactions
Time Clock Capabilities
Records results for each wire of a dual wire wrist strap
Records results for the left and right footwear
Wrist / Foot symbols make it easy for operators to identify what test to perform
All changes are recorded and viewed through the server software
The software features automatic saving function, which can save 300,000 testing records in case of an unexpected shutdown.
Simultaneously managing multiple ESD test access control systems through the TCP/IP network. There is stand-alone software (for free) and
the online software (charges required). The stand-alone version can only run on one stand-alone computer for monitoring and management,
while the online version can be authorized to multiple computers for monitoring and management.
The employees can be grouped based on permission for real-time monitoring or batch management, which is user-friendly.
Displaying the test results of employee daily, and saving them in the files, which can be searched out, saved and analyzed.
Displaying the test results and recording the causes for the errors, such as failure of ESD wrist strap, left shoe, right shoe, or unauthorized
cards, and displaying the department, door number, card number, job number, name and other information.
Searching by various criteria, such as a month, date, department, employee or various causes of errors to generate the multi-functional ESD
access control management sheets.
Capable of converting Excel employee information sheets. The existing employee information sheet in Excel format can be directly imported
into the ESD software,where the permissions can be automatically uploaded and authorized.
With optional attendance management software, with which the ESD test records can be used directly in attendance management.
We can provide communication ports for customer’s redevelopment, or we can re-develop the software as required by customers to satisfy
This software can be used for remote management, which makes the software maintenance more flexible and convenient.
Compatible with all Windows operating systems.
Microsoft Windows XP or more senior Windows system
500 megahertz (MHz) processor
2G internal memory
128G hard disk
Broadband connection, 128 kilobits per second (Kbps)
Microsoft Excel 2007
Microsoft Access 2007
Desktop and server installation packages available
regular basis. Use 99% isopropyl alcohol to clean the foot plate and test
switch. Other cleaners are susceptible to leaving residue on these
NOTE: Avoid wetting or mopping the foot plate. This may yield false test
results and cause damage to its internal circuitry
Technical Data Sheet
Input Voltage and Frequency (External Adaptor): AC / DC Power Adapter
Power Input: 100-240VAC, 50/60 HZ
Power output: 12VDC, 0.5A
Cable Length: 1.5m
Operating Temperature: 21℃-30℃ for 1 Gigohm test limit // 5℃-30℃ for all other test limits
Test Accuracy: ±15% for 1 Gigohm footwear test limit // ±10% for all other test limits
Test Switch Voltage: 28VDC @ open circuit
Wrist Strap and Footwear Test Voltage: 28VDC @ open circuit // (Test current is limited by resistors and varies
on the test range setting (100K-1G)
ESDMAN6809 Dual Independent Foot Plate:
Dimensions: 49cm x 39cm x 2.5cm
Specification_ESD Access Control System_OK.pdf
ESDMAN 6809_Software Installation and Network Debugging_OK.pdf
Template_User File of ESD Access Control System.xlsx